InnoSpire Elegance Compressor Nebulizer System is a home tabletop system that includes everything you need for fast, reliable delivery of medication. The unit's compact, lightweight design means you can move it to whichever room you like, or take it along on your travels. Just plug it in, add your medication to the included SideStream nebulizer and this ready-to-use device will do the rest, quickly and efficiently. InnoSpire Elegance proudly comes from the InnoSpire family of compressor nebulizer systems by Philips Respironics. 5-Year Manufacturer's Warranty.
What Comes with the InnoSpire Elegance?
Specifications for Respironics InnoSpire Elegance Compressor Nebulizer System
|• Weight:||3 lbs|
|• Dimensions:||4.2" x 6.5" x 6.5"|
|• Electrical Rating/Power:||120V/60Hz, 1.7 Amp|
|• Nebulization Capacity:||8 ml|
|• MMAD:||(80%) <5 microns|
|• Treatment Time:||5-7 minutes|
|• Maximum Pressure:||44 psi|
|• Maximum Flow Rate:||9.4 LPM|
|• Operating Flow Rate:||7 LPM @ 10 psi|
|• Mode of Operation:||Continuous|
|• Compliance:||International safety and essential performance standard IEC 60601-1, third edition and its amendments|
Directions for Use:
Please see the Owner's Manual tab for full instructions on the Respironics InnoSpire Elegance Compressor Nebulizer. Also on that tab is an Owner's Manual for the SideStream Disposable Nebulizer that comes with the device, plus one for the optionally available SideStream Reusable Nebulizer.
The InnoSpire Elegance comes equipped with a cylindrical, white air filter to keep dust and other particles out of the device. Check this filter regularly by simply pulling the filter out from the right side of the device. Replace every 30 days, or sooner if discolored or wet.
Please refer to the Owner's Manual(s) for detailed cleaning instructions.
Just use a slightly dampened cloth to wipe the outside of the unit.
Clean after every use. Disconnect tubing from the nebulizer and wash all parts (except tubing) separately in mild liquid detergent and warm water. Do not wash the tubing. Rinse thoroughly, shake out excess water, and allow parts to air dry before reassembling. Can also be washed once a week in dishwasher (top rack).
Disinfect once a week (reusable only). Disconnect tubing from the nebulizer and boil all the parts (except tubing) in water with a few drops of dish soap for 10 minutes. Do not disinfect the tubing. Rinse thoroughly, shake out excess water, and allow to air dry before reassembling.
Below, for your convenience, please find the Respironics Part #s, Descriptions, and NDC and HCPCS Codes. For information about Medicare and Medicaid reimbursement, please visit www.cms.gov. For information about reimbursement from private health plans, please contact your health insurer or plan administrator.
|Part #||Description||NDC Code||HCPCS Code(s)|
|1100134||InnoSpire Elegance with SideStream Disposable Nebulizer||08373-0134-00||E0570|
|HS800||SideStream Disposable Nebulizer||08373-0008-00||A7003|
|HS860||SideStream Reusable Nebulizer||08373-0860-00||A7005|
Click here to view the Owner's Manual for the Respironics InnoSpire Elegance Compressor Nebulizer System.
Click here for a Basic Guide on using a Respironics compressor (tabletop models) with a SideStream nebulizer.
Please click on a link below to view the Owner's Manual for each of the Accessories available at our site for the Respironics InnoSpire Elegance Compressor Nebulizer System.
We apologize. While we do offer the SideStream Pediatric Mask as an optional accessory for the InnoSpire Elegance Compressor Nebulizer System, we do not have an Owner's Manual for this item. The mask easily attaches to the SideStream Nebulizer that comes with the system by using its swivel connector. The mask is contoured to comfortably fit the face, and is recommended for children ages 5 years and younger. For full instructions on using and caring for the reusable mask, please refer to the leaflet that comes in the package.